JL VJT 2723e
Refrigerator of 141 l; adjustable temperature from 2°C to 15°C; glass door; 3 shelves; alarms: under and over temperature, door open and power failure; lockable door. Suggested use: universal.
5 Item Items
Warning: Last items in stock!
|Adjustable temperatures (°C)||1°C...15°C|
|Exterior Dimensions mm WxDxH||600x615x830|
|Inner Dimensions mm WxDxH||440x435x670|
|Inner Gross/Net Capacity Liters||140/130|
|Shelves / Baskets||3|
|Inner Fan||Yes, disconnectable|
|Control Panel||Display LCD|
|Open door alarm||Yes|
|Power Failure Alarm||Yes - it last about 12 hrs.|
|Visual and acoustic temperature alarm||Yes - Min. & Max temperature|
|Adjustable alarms||As an option|
|Max Ambient Temp. to reach the Min. Temperature||38|
|Porthole for PT 100||As an option|
|Noise level (db)||45|
|Max Power Input (W)||80|
|Energy consumption [kWh/24h]||0.898|
|Connection rating (A)||1|
|Supply Voltage (V) / Frequency (Hz)||220 - 240 / 50|
|Comply with Italian Republic Farmacopea Ufficiale XII edizione Tabella VI||Yes|
|Various||Sicurezza blocco tasti, ventilato, porta con cerniera a destra reversibile, chiusura porta automatica|
|Warranty||International Liebherr Warranty 24 Months|
Refrigerators efficient and in compliance with the pharmacopoeia. The backlit LCD display constantly monitors the internal temperature and reports any alarms over and under temperature if should they persist.
Refrigerators powerful and well insulated because suitable for ambient temperatures from 10 ° C to 43 ° C; housing in white painted steel; internal space in PS.
LCD (Liquid Crystal Display) with adjustable brightness
Low and high temperature alarms and door open;
Safety key lock;
Reversible door with right hinge;
Automatic door locking
Internal LED light
VJT 2723e Test
Tests with a control probe in liquid. Red line = room temperature; green line = temperature in the medium.
Set: 5 ° C.
The test shows a rapid drop in temperature to reach 5 ° C (in the sample) in 75 min.; the temperature stabilizes between 4.4 and 4.6 ° C with no major shifts.
Test is generic with a datalogger not certified but still subjected to comparison with certified thermometer in our laboratories. The behavior is typical of a VJT 2723, but is not representative of the entire production. The accuracies may vary from model to model and depending on the thermal differences and load.